Sector mass spectrometer
| IUPAC RECOMMENDATIONS 2013
K. K. Murray, R. K. Boyd, M. N. Eberlin, G. J. Langley, L. Li and Y. Naito, Pure Appl. Chem., 2013, 85, 1515-1609, 10.1351/PAC-REC-06-04-06.
|Sector mass spectrometer|
|Mass spectrometer consisting of one or more magnetic sectors for recording a mass spectrum of a monoenergetic beam of ions. Such instruments may also have one or more electric sectors for energy dispersion that can be arranged for velocity-focusing by compensating for dispersion by the magnetic sector if the beam is not truly monoenergetic. Such a combination provides double focusing of ions of the same m/z, i.e., focusing with respect to both direction and speed.|
A. O. Nier. Int. J. Mass Spectrom. Ion Processes 100, 1 (1990). (http://dx.doi.org/10.1016/0168-1176(90)85063-8 )
|Definitions of Terms Relating to Mass Spectrometry (IUPAC Recommendations 2013); DOI: 10.1351/PAC-REC-06-04-06 © IUPAC 2013.|